The computerized instrument tends to be more sensitive to blood pressure changes than its analog equivalent. For this reason we recommend massaging the cuff twice and running at around 70mm Hg. Examiners usually achieve the best results by inflating to 90mm Hg and massaging the cuff down to 70mm Hg, then reinflating to 80mmHg and massaging back down to 70mm Hg. Always remember the lower your sensitivity and higher the pressure (within limits) the better the tracing.
This Article has been viewed 1464 times
Question Last Updated on: 6/7/2007
Unable find an answer to the question that you looking for in our FAQ section? Submit your question directly to our customer service representatives.
Click here to view the status of and reply to the questions you have submitted.